Function Command String (FCS / MLB) for SMP Controllers
Sheffield SMP controllers accept not only traditional commands from CMM-Manager program execution but also a special command set unique to Sheffield SMP controllers. It is possible to create aubile, warnings, pause program execution, toggle Joystick / Auto Mode, etc. by sending FLB / FCS commands to controller during program execution.
From the Program Ribbon Tab, press the Function Command String button, then type FLB / FCS command into the text entry, then click OK. When program runs, this FCS will be executed. See attached FLB / FCS reference for details on available FCS commands.
Sheffield FCS / FLB Commands
Sheffield - MP / SMP (Manual)
Models: MP30/35/30-II Connection: Serial / RS-232 *null modem Default settings: 9600, O, 8, 1 Some DCC capable controllers use a baud rate of 19200. After a change of baud rate from CMM-Manager Machine Setup, the controller has to be restarted. MP35 ...
Sheffield - SMP / UMP (DCC)
Models: SMP-300, SMP-350, SMP-400, UMP-360 Important! - Backup MP Configuration Connection: Serial / RS-232 *null modem Default settings: 19200, O, 8, 1 (SMP-300/350) 57600, O, 8, 1 (SMP-400) Scanning machines with SMP-400 can communicate at either ...
Possible CMM-Manager Installer Failures and Resolution
On occasion, customers may encounter issues with the installation package not completing as intended. Depending on the circumstances, it is possible that the CMM-Manager application can still be launched and used without realizing any immediate ...
Troubleshooting Newmark Rotary Indexer
If your rotary indexer has no encoder, it is possible that the table will rotate a couple of degree and stop irrespective of the requested command. When this happens, follow the instructions listed below to troubleshoot this issue: Uncheck the ...